|NOMINAL TORQUE||0.5 – 2.5 – 5 – 10- 25 – 50 N•m
100 – 250 – 500 Nm
1000 – 3000 5000 Nm
|LINEARITY and HYSTERESIS||£ ± 0.2 %|
|TEMPERATURE EFFECT (1°C):
a) on zero
b) on sensitivity
£ ± 0.02 %
£ ± 0.02 %
| NOMINAL POWER SUPPLY
MAX. POWER SUPPLY
ROTATING torque transducer WITHOUT Contact transmission
The dynamometer was designed to measure static and dynamic torque on rotating machines, test benches and automatic clamping systems.
The dynamometer measures the HOURLY torque with a POSITIVE output in voltage or the ANTICLOCKWISE torque with the NEGATIVE output in voltage.
For correct operation and measurement accuracy, it is necessary to prepare:
2 SHAFT CONNECTIONS with bellows or disc pack according to nominal torque and rotation speed (max 4000 rpm).
1 ADJUSTABLE SUPPORT which allows during installation, to align the sensor with the two connecting shafts (tolerance ± 0.1 mm).
The coupling of the couplings to the dynamometer must be disconnected from the machine (system) with the dynamometer connected to the display, thus verifying in real time that no unwanted torque, bending and tension are generated with possible overload of the dynamometer.